A Complete HIOKI Digital & Analog Clamp Tester /A AC .. CARRYING CASE ×1. Hand strap ×1. Instruction manual ×1. Hioki Pdf User Manuals. View online or download Hioki Instruction Manual. View and Download Hioki instruction manual online. Clamp On AC/DC HiTester. Measuring Instruments pdf manual download.
|Published (Last):||5 February 2010|
|PDF File Size:||19.99 Mb|
|ePub File Size:||12.1 Mb|
|Price:||Free* [*Free Regsitration Required]|
Using a one-chip microprocessor, the Set a range based on the unit’s measurement range and other instruments, such as recorders. Remove the rear cover. Check the Unit and the Supplied Accessories Press to display hioko.
Select a proper mode according to the waveform shown below: Would you prefer to visit our US mwnual www. FAST Instead, the unit symbol blinks.
HIOKI INSTRUCTION MANUAL Pdf Download.
Ask an Expert More Hioki Products. Also See for Instruction manual – pages. This product includes one lens. Ask our experts To ask our Hioki experts a question please include your contact details.
Remove the rear cover and insert a battery. Inrush current hiomi value. Always switch to the range you will use a manual range before performing the adjustment. Using a Hioki Current Clamp Meter: We also accept wire transfers, cheques and purchase orders on approved credit.
To print the manual completely, please, download it.
Hioki 3284 Instruction Manual
The torque about 0. The auto power-off function is automatically disabled.
Save my login information. Remove the two fastening screws of the rear cover, using a Phillips screwdriver.
Just Added to Your Cart: Replace the battery to ensure accuracy. Comments to this Manuals Your Name.
Request for Quote To manuwl an official quote for the Hioki please include your quantities and contact details. Not finding what you’re looking for? Every time one minute elapses, one segment of the flashing bar graph goes on.
Got it, continue to print.
Hioki CLAMP ON AC/DC HiTESTER
Be the first to know when we have news, discounts, special offers and promotions. The counter update may be altered according to measurement conditions. There isn’t an annunciator ” “. Next, insert the red test lead to V and the black test lead to COM of the voltage measurement terminal.
Chapter 2 Measurement Procedure Explains how to use the for measurement. We accept these payment methods: In the manual, the symbol indicates particularly important information that the user should read before using the instrument. Page 5 We have tried to bring this manual as close to perfection as we could achieve. Symptom Battery Battery clip Test leads Unit does not come indication appears and unit immediately turns off.
When all segments on the bar graph go on, the elapsed time is 30 minutes. Then the model name is shown, and the bar graph indicates the battery condition. It is recommended that a new battery should be used, when the blackout seems to be short.